The Simpson 160 is only available in new shop sets
as a substitute for the TS-352B/U or AN/URM-105.
b. Using the Ohms Scale. The ohms scale is
used to make tests for continuity, shorts and re-
(1) Zeroing the Meter (Fig. 4-10). The
multimeter must be set up and zeroed before
making these tests. Do the following steps that
match the multimeter.
If the needle will not zero, replace the batteries. If
the needle still will not zero after replacing the
batteries, turn the meter in for repair.
Zeroing the multimeter (sheet 1 of 3).
4 - 1